View and Download Dixell XR20C installing and operating instructions online. Digital controller with off cycle defrost. XR20C Controller pdf manual download. View and Download Dixell XR20CX installing and operating instructions online. Digital controller with off cycle defrost. XR20CX Thermostat pdf manual. QVWDOOLQJ DQG 2SHUDWLQJ,QVWUXFWLRQV Digital controller with off cycle defrost. SET: To display target set point; in programming mode it selects a.
|Genre:||Health and Food|
|Published (Last):||26 January 2018|
|PDF File Size:||12.96 Mb|
|ePub File Size:||15.82 Mb|
|Price:||Free* [*Free Regsitration Required]|
Model XR20C, format 32 x 74 mm, is a digital thermostat with off cycle defrost designed for. The digital input can operate as. It provides a relay output to drive the. Page of 4 Go. In case of fault in the thermostat probe the start and stop of the compressor are timed through. The instrument must not be. Comments to this Manuals Your Name. XR20CX panel mounting in a 71x29mm panel cut-out Protection: Push and immediately release the SET key: Ptb Map code Dixell S.
Check the supply voltage is correct before connecting the instrument.
The “Hi” message will be displayed followed by the maximum temperature recorded. Defrost is performed through a simple stop of the compressor. Push the SET key for more than 2 seconds to change the Set point value. To see the max. DOWN To see the min stored temperature; in programming mode it browses the.
Dixell XR20CX Installing And Operating Instructions
Fit the probe where it is not accessible by the End User. Enter text from picture: This manual is part of the product and should be kept near the instrument for easy and. Push and immediately release the SET key or wait for 5 seconds to.
Consider the maximum current which can be applied to each relay see Technical Data. Each LED function is described in the following table. The instrument is fully configurable through special parameters that can be easily. Hold press the SET key for more than 3s, while the max. The instrument shall not be used for purposes different from those described hereunder.
The “Lo” message will dixelll displayed followed by the minimum temperature recorded. Ensure that the wires for probes, loads and the power supply are separated and far. Instrument XR20CX shall be mounted on vertical panel, in a 29×71 mm hole, and fixed using the special bracket supplied. In case of failure or faulty operation send the instrument back to the distributor or to. Got it, continue to print. Installing and Operating Instructions.
By pressing ‘print’ button you will print only current page. Add to my manuals Add. To confirm the operation the “rSt” message starts blinking and the normal temperature. Diexll case of applications in industrial environments, the use of mains filters our mod.
To memorise the new set point value push the SET key again or wait 10s. Don’t show me this message again.
XR20C – AS Catering Supplies
To display target set point; in programming mode it selects a parameter or confirm. Don’t have an account? To print the manual completely, please, download it. CCS Set point for continuous cycle: To change the Set value push the 9 or 8 arrows within 10s.
By pressing the 8 key again or by waiting sixell the normal display will be restored. Page 4 Ptb Map code Dixell S. By pressing the 9 key again or by waiting 5s the normal display will be restored. Parameter “IdF” controls the. Do not expose to water or moisture: Push the DEF key for more than 2 seconds and a manual defrost will.
Page 2 Energy Saving cycle.